The MAX30001 is a complete, biopotential and bioimpedance (BioZ), analog front-end (AFE) solution for wearable applications. It offers high performance for clinical and fitness applications, with ultra-low power for long battery life. The MAX30001 is a single biopotential channel providing electrocardiogram (ECG) waveforms, heart rate and pacemaker edge detection, and a single bioimpedance channel capable of measuring respiration.
The biopotential and bioimpedance channels have ESD protection, EMI filtering, internal lead biasing, DC leads-off detection, ultra-low-power, leads-on detection during standby mode, and extensive calibration voltages for built-in self-test. Soft power-up sequencing ensures no large transients are injected into the electrodes. Both channels also have high input impedance, low noise, high CMRR, programmable gain, various low-pass and high-pass filter options, and a high resolution analog-to-digital converter. The biopotential channel is DC coupled, can handle large electrode voltage offsets, and has a fast recovery mode to quickly recover from overdrive conditions, such as defibrillation and electro-surgery. The bioimpedance channel includes integrated programmable current drive, works with common electrodes, and has the flexibility for 2 or 4 electrode measurements. The bioimpedance channel also has AC lead off detection.
The MAX30001 is available in a 30-bump wafer-level package (WLP), operating over the 0°C to +70°C commercial temperature range.
|Shutdown Mode Supply Current|
|247||0.42||1.1||2||18||0 to +70|
|See All Bio-Sensors (15)|
| Pricing Notes: |
This pricing is BUDGETARY, for comparing similar parts. Prices are in U.S. dollars and subject to change. Quantity pricing may vary substantially and international prices may differ due to local duties, taxes, fees, and exchange rates. For volume-specific and version-specific prices and delivery, please see the price and availability page or contact an authorized distributor.
|Device||Fab Process||Technology||Sample size||Rejects||FIT at 25°C||FIT at 55°C|
|DesignSoln||7157||How to Add ECG to Your Wearable Without Raising Your Heart Rate|
|App Note||6952||How to Measure Biopotential ECG Using a Chest Strap|
|App Note||6951||Take a Weight Off Your Chest with a Wrist-Worn ECG Monitor|
|User Guide||6780||MAXREFDES101# Health Sensor Platform 2.0 User Guide|
|Tutorial||6778||How to Measure Biopotential ECG Using a Chest Strap|