Keywords: blood pressure, systolic, diastolic, heart rate monitor, SpO2, blood oxygen, blood oxygen level, wearable, HRM,sensor, sensormodule, PPG, raw data, sensor hub, SPI, I2C,\r\ncalibration Related Parts
USER GUIDES & MANUALS 6921
Measuring Blood Pressure, Heart Rate, and SpO2 Using MAX32664D
Abstract:The MAX32664D is a variant of the MAX32664 sensor-hub family, which is specifically targeted for finger-based measurement of blood pressure, heart rate, and SpO2. Combined with the MAX30101 pulse oximetry and heart-rate monitor module and powered by Maxim's BPT algorithm, it provides sensors’ raw data, as well as calculated systolic and diastolic blood pressure, SpO2, and heart-rate data to a host device through its I2C slave interface. This document provides step-by-step instructions that enable a user to communicate with the MAX32664D, and to calibrate, configure, and receive measurement and monitoring data.
Thank You for interest in Maxim Integrated. Our free samples program limits the quantities that we can provide to each customer per calendar year.If you feel that you have received this message in error, please contact email@example.com. Alternatively, you can click here to place an order.
Request a Quote
Your quote has been successfully submitted. You will receive an email within 24 hours with pricing and availability. Visit My Quotes Thank you!