How to Measure Biopotential ECG Using a Chest Strap
By: John DiCristina
Abstract:This tutorial examines key design considerations for reliable biopotential ECG measurement in wearable ECG chest-strap applications such as fitness heart-rate sensors. Topics include chest-strap electrode selection, rejection of ECG motion artifacts, and how to maintain a low-power profile. The benefits of a fully integrated biopotential sensor solution are reviewed and examples of available design and development resources are provided.
Thank You for interest in Maxim Integrated. Our free samples program limits the quantities that we can provide to each customer per calendar year.If you feel that you have received this message in error, please contact firstname.lastname@example.org. Please click here to place an order.