Build Low Cost White Noise Generator
This white noise generator is based on the avalanche noise generated by a zener breakdown phenomenon. It is created when a PN junction is operated in the reverse breakdown mode. The avalanche noise is very similar to shot noise, but much more intense and has a flat frequency spectrum (white). The magnitude of the noise is difficult to predict due to its dependence on the materials.
Basically the circuit has 2 portions: a zener diode, and MAX2650 Low Noise Amplifiers (LNA). The diode is reverse-biased and noise output is taken from the anode. The MAX2650 is a wide-band Low-Noise-Amplifier with 19dB gain operating from DC to 1GHz.
Summary of Test ResultsThe white noise generator circuit is realized with 2 boards, one for the zener diode, one for the MAX2650 LNA. The output trace from the anode of the zener to the input of the LNA is designed to be 50Ω. The output of the LNA is measured by a Rhode & Schwarz Spectrum Analyzer. For practical purpose, noise power of -60dBm or above is desired within the frequency band of interest.
After several experiments, I obtained the following results:
- The noise output power is mostly decided by the diode's breakdown voltage. I have used zener diodes with breakdown voltage of 4V(1N749), 5.1V(1N751), 5.6V(1N752), 7.5V(1N755), and up to 12V(1N759). 12V zener generated noise power is at least 15~20dB higher than that generated by 5V zener.
- Even with 12V zener, 20dB gain from the LNA is not enough to produce noise power higher than
-60dBm.We need at least two LNAs in cascade (38~40dB gain).
- The output noise power is almost independent of the source current. With the potentiometer, the current into the diode can vary from 0mA to 100mA. Within this source current range, the noise power varies pretty randomly within ±1dB. It seems in zener diode breakdown phenomena, the avalanche noise dominates over other noise source, such as shot noise (which is proportional to current), flicker noise and thermal noise.
- The frequency/power spectrum of the output noise is not perfectly flat over a wide frequency range. As shown in test result Figure 2, from 1MHz to 100MHz, noise power drops almost 10dB. This could be because of higher on-board attenuation for higher frequency components.
- 2 LNAs were used in cascade for the white noise generator, which produced 38dB~40dB gain.
Figure 1. White-noise generator circuit.
Figure 2. White-noise generator output spectrum.
ConclusionIn this article, a design idea of a White Noise generator is realized The test result shows a pretty wide-band white noise generated with several inexpensive components.
A similar version of this article appeared in the September-October 2004 issue of Electronica magazine.